High-Accuracy Wet-Gas Multiphase Well Testing and Production Metering
- David I. Atkinson (Schlumberger) | Oyvind Reksten (3-Phase Measurements A/S) | Gerald Smith (Schlumberger) | Helge Moe (Framo Engineering A/S)
- Document ID
- Society of Petroleum Engineers
- SPE Journal
- Publication Date
- June 2006
- Document Type
- Journal Paper
- 199 - 205
- 2006. Society of Petroleum Engineers
- 4.4.3 Mutiphase Measurement, 5.3.2 Multiphase Flow, 1.10 Drilling Equipment, 4.3.4 Scale, 4.1.5 Processing Equipment, 5.3.4 Integration of geomechanics in models, 4.6 Natural Gas, 5.6.4 Drillstem/Well Testing, 4.1.2 Separation and Treating
- 0 in the last 30 days
- 633 since 2007
- Show more detail
- View rights & permissions
|SPE Member Price:||USD 5.00|
|SPE Non-Member Price:||USD 35.00|
Dedicated wet-gas flowmeters are now commercially available for the measurement of gas and liquid flow rates and offer a more compact measurement solution than does the traditional separator approach. The interpretation models of traditional multiphase flowmeters emphasize the liquid rate measurements and have been used to well test and meter mostly liquid-rich flow streams. These models were not developed for the measurement of gas flow rates, particularly those of wet gas.
A new interpretation is described that allows a traditional multiphase flowmeter to operate in a dual mode either as a multiphase meter or as a wet-gas meter in 90 to 100% gas. The new interpretation model was developed for a commercially available multiphase flowmeter consisting of a venturi and a dual-energy composition meter. This combination results in excellent predictions of the gas flow rate; the liquid rate prediction is made with acceptable accuracy and no additional measurements.
The wet gas and low-liquid-volume-fraction interpretation model is described together with the multiphase flowmeter. Examples of applying this model to data collected on flow loops are presented, with comparison to reference flow rates. The data from the Sintef and NEL flow loops show an error (including the reference meter error) in the gas flow rate, better than ± 2% reading (95% confidence interval), at line conditions; the absolute error (including the reference meter error) in the measured total liquid flow rate at line conditions was better than ± 2 m3/h (< ± 300 B/D: 95% confidence interval).
This new interpretation model offers a significant advance in the metering of wet-gas multiphase flows and yields the possibility of high accuracies to meet the needs of gas-well testing and production allocation applications without the use of separators.
There has been considerable focus in recent years on the development of new flow-measurement techniques for application to surface well testing and flow-measurement allocation in multiphase conditions without separating the phases. This has resulted in new technology from the industry for both gas and oil production. Today, there are wet-gas flowmeters, dedicated to the metering of wet-gas flows, and multiphase meters, for the metering of multiphase liquid flows.
The common approach to wet-gas measurement relates gas and liquid flows to a "pseudo-gas flow rate" calculated from the standard single-phase equations. This addresses the need for gas measurement in the presence of liquids and can be applied to a limit of liquid flow [or gas volume fraction, (GVF)], though the accuracy of this approach decreases with decreasing GVF. The accurate determination of liquid rates by wet-gas meters is restricted in range.
The application and performance of multiphase meters has been well documented through technical papers and industry forums, and after several years of development is maturing (Scheers 2004). Some multiphase measurement techniques can perform better, and the meters provide a more compact solution, than the traditional separation approach. It is not surprising that the use of multiphase flowmeters has grown significantly, the worldwide number doubling in little over a 2-year period (Mehdizadeh et al. 2002). Multiphase-flowmeter interpretation emphasizes the liquid rate measurement, and the application of multiphase flowmeters has been predominantly for liquid-rich flow stream allocation and well testing.
|File Size||1 MB||Number of Pages||7|
Al-Taweel, A.B., Barlow, S.G., and Hussain, S. 1998. Wellsite Wet Gas Measurement Systemin Saudi Arabia. Paper SPE 49162 prepared for presentation at the SPEAnnual Technical Conference and Exhibition, New Orleans, 27-30 September.
Atkinson, D.I., Berard, M., Hanssen, B-V., and Segeral, G.1999. NewGeneration Multiphase Flowmeters from Schlumberger and Framo Eng. AS.Proceedings of the International North Sea Flow Measurement Workshop. EastKilbride, Scotland: Natl. Engineering Laboratory.
Atkinson, D.I., Berard, M., and Segeral, G.2000. Qualification of a NonintrusiveMultiphase Flowmeter in Viscous Flows .Paper SPE 63118 presented atthe SPE Annual Technical Conference and Exhibition, Dallas, 1-4 October.
Chisholm, D. 1977. Two-Phase Flow Through Sharp Edged Orifices. J.Mechanical Engineering Sciences 19(3):128.
de Leeuw, R. 1997. Liquid Correction of Venturi Meter Readings in Wet GasFlow. Proceedings of the North Sea Flow Measurement Workshop. East Kilbride,Scotland: Natl. Engineering Laboratory.
Intl. Organization for Standardization. 1991. 5167-1:1991(E): Measurement offluid flow by means of pressure differential devices. ISO: Geneva,Switzerland.
Konopczynski, M.R, and de Leeuw, H.: Large-Scale Application of Wet-GasMetering at the Oman Upstream LNG Project . Paper SPE 63119 presented atthe 2000 SPE Annual Technical Conference and Exhibition, Dallas, 1-4October.
Lin, Z.H. 1986. Encyclopedia of Fluid Mechanics. Vol. 3. Gulf PublishingCo.: Houston.
Lund, J.S., Tait, A.R., and Clark, S. 1999. A Wet Gas Meter forGas/Condensate Flow Measurement. Proceedings of the International Conference onMultiphase Flow. 303.
Mehdizadeh, P., Marrelli, J., and Ting, V.C. 2002. Wet Gas Metering: Trends inApplications and Technical Developments. Paper SPE 77351 presented at theSPE Annual Technical Conference and Exhibition, San Antonio, Texas, 29September-2 October.
Murdock, J.W. 1962. Two-Phase Measurements With Orifice Plates. Journal ofBasic Engineering. Trans ASME:567.
Nederveen N., Washington, G.V., and Batstra, F.H. 1989. Wet Gas Flow Measurement. PaperSPE 19077 presented at the SPE Gas Technology Symposium, Dallas, 7-9 June.
Pinguet, B.G. and Barreto, W. 2004. Multiphase Flowmetering FieldExperiences in Brazil-An Artificial Lift Focus. Proceedings oftheInternational SE Asia Hydrocarbon Flow Measurement Workshop. EastKilbride, Scotland: Natl. Engineering Laboratory.
Scheers, L. 2004. Multiphase and Wet Gas Flow Metering—On Its Way FromNursing to Mature Technology. Proc., Multiphase Measurements User Roundtable.Texas A&M U.: College Station, Texas.
Smith, R.V. and Leang, J.T. 1982. Evaluations of Correlations for Two-PhaseFlowmeters, Three Current—One New. J. of Engineering for Power. Trans. ASME97(4):683.
Steven, R.N. 2002. Wet Gas Metering With aHorizontally Mounted Venturi Meter. Flow Measurement and Instrumentation12(5-6):361.